I bought DW1000 evaluation KIT. yesterday I have a problem with 1 of the anchor node, suddenly the LCD shows message “ERROR INIT FAIL” in the middle of experiment.
I try to find the problem from the user manual “EVK1000 USER MANUAL”, it said that we should make sure the switch 2(S2) is all ON position, but its not solve the problem. I try to change the DIP switch from ON to OFF then back to ON again several times, but still error.
Do you know why this error happens? Because the error message is appear suddenly, and we did not change the configuration from the beginning.
forget to mention, my Decawave version still V2.10. Do I need to update the software?